Maintaining the Outside Temperature Humidity Sensor in VP2 will assure that your weather instruments will continue to perform at their highest accuracy.
These are some issues you may encountered if your not maintaining the Outside Temperature Humidity Sensor in VP2.
- Dust build-up can interfere with air flow over the sensor.
The common solutions you may used in maintaining the anemometer wind vane is the following.
- When you remove the plate in the radiation shield on which the the temp-hum board is installed, you will see a small, white, vented cover lined with gold-colored mesh.
- Using a clean, dry toothbrush, gently remove any dust from the cover. You can remove the cover from the board and take out the mesh to do a more thorough cleaning if necessary
Now that is all the information you may need in Outside Temperature Humidity Sensor in VP2. For any question you may have, just visit the contact form and don’t forget to hit the send.